The 589 saccharimeter from Rudolph Research Analytical provides the same features, measuring speed and accuracy of the AUTOPOL 880 but at the single wavelength of 589 nm.
Whether your sugar analysis requires a 25mm, 50mm, 100 mm or 200 mm cell, whether you use 1/2 normal (13g/100ml), 1/4 normal (6.5g/100ml) or full weight sugar solutions, the AUTOPOL® 589, 880, 880T, and 880 PLUS saccharimeters provide the correct multiplier to display the reading as if the sample were prepared as a 1 normal solution in a 200 mm optical pathlength cell.
Calibrating the AUTOPOL® is quick and simple. First, place a Rudolph Quartz Control Plate calibration standard on the optical rails and enter the Quartz Plate value in the appropriate window. Second, enter the temperature to correct the measurement to either the ICUMSA standard temperature of 20°C or the Tropical reference temperature of 27.5°C. Third, press 2 buttons and calibration is complete. (See Technical Bulletin 913 for more details on Rudolph Calibration Standards).
All AUTOPOL® Digital Saccharimeter models have a temperature probe in the measurement compartment. The probe can be inserted into all 14 style, 40T style, 41T style and 33 style measurement cells.
Rudolph knows Sugar Technologists around the world demand 24 hour operation, 7 days a week for up to 9 months out of the year. They expect 0.02 °Z reproducibility and accuracy. Rudolph Research understands these demands and requirements, because Rudolph has been serving the sugar industry since 1940. Throughout those sixty plus years, Rudolph has dedicated itself to manufacturing quality instrumentation and providing superior technical and service support, both directly and through its worldwide dealer network. This dedication is consistent with Rudolph’s over 5,000 instruments installed worldwide, with many of those instruments still in operation after 25 years of service.